Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||24 November 2016|
|PDF File Size:||14.61 Mb|
|ePub File Size:||8.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
With the A in its turn-on condition, set the manuaal to 10 MHz, function to square wave, and amplitude to. The lines should not be separated by more than ten divisions on the display.
Option is a temperature stabilized 10 MHz oscillator which provides improved frequency stability see specifications in Table Valid data for SM is: Multimeter will indicate between 2. General Interface Management Lines. If no data has been stored in a register, the recall command for that register will be ignored. Verify that all harmonics are at least 65dB below the fundamental.
Amplitude Limits of AC Function. Min, Sine 10 V 1 mV 3. Return CRO sweep vernier to calibrate and set main sweep to. These keys select the output signal function or dc only see Paragraph Np the original packaging has been retained, pack the instrument in the same manner as it was received. These instruments do not have CR An external reference may be used to phase Jock the internal 30 MHz reference see Paragraph Set the oscilloscope controls as mahual Press the key in the lower right corner of the A front panel to select the High Voltage output.
Remove the filter and wash thoroughly with soapy water, rinse elcan, and air dry.
HP A CALIBRATION MANUAL Pdf Download.
TB 4 Enter Printer records the results of the serial poll. Adjust the dc power supply janual a digital voltmeter reading of In standard instruments, switches signal output from front to rear panel and vice versa.
Change A frequency to 1 MHz. Rear panel output is active when the annunciator in the center of the key is on.
HP 3325A Synthesizer/Function Generator Operating and Service Manual
No adjustments can be made. If shipping to a Hewelett-Packard office or service center, attach a tag indicating the type of service required, return address, model number, and full serial number. X Drive Ramp Output. Set the A as follows: Verify that this spur is at least 70dB below the fundamental.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
This procedure checks the square wave output for frequency, rise time, and abberrations. The dc offset is also checked in these tests.
Auxiliary Output Sine Function Only See Service Group C. Triangle and Ramp Linearity Test.
This TTL Compatible output goes low at the selected marker frequency during a sweep up, and high at completion of the sweep see Paragraph These bits are the same for both listen and talk addresses, and the sixth and seventh bits determine whether hl address is listen 01 or talk Page 8 TB a 50 Hz. Rise time should be less than 20 nanoseconds.
These instruments do not, however, have CR TB a 50 Hz. If the oscilloscope is an -hp- Model A, set the input switch to the ohm position. The oscilloscope should display one triangle wave with no visible irregularities in either slope. Any equipment that satisfies the critical specifications given in the table may be substituted for the recommended model.
Connect the A signal output to the electronic counter channel A input with a 50 Q load.