Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual. Operating and Service Manual. HP A. Transmission Impairment. Measuring Set. Serial Numbers. This manual applies to instruments with serial numbers. HP / Hewlett Packard / Agilent Used Oscilloscopes, Analog and Digital, Power Supplies, AC DC Power Supplies, Single Output, Multiple Dual Triple Quad.
|Published (Last):||22 December 2012|
|PDF File Size:||7.57 Mb|
|ePub File Size:||18.20 Mb|
|Price:||Free* [*Free Regsitration Required]|
Performance Tests Model Page 15 Model Page 22 Installation Table There will be no values displayed hl in this configuration the transmitter is turned off and a quiet termination is supplied to the TRMT jack. Indented primarily for field service installation and maintenance, the A weighs only three pounds with batteries.
You will find manua other available. Packaging should be the same as what is found in a retail store, unless the item is handmade or was packaged by the manufacturer in non-retail packaging, such as an unprinted box or plastic bag.
HP A Operating & Service Manual | eBay
Please note that Hewlett. I,rvel can be read in the left display and frequency in the right. Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc.
Accepts power cord supplied with instrument. Page 28 Operation Model Always connect power cord to a properly grounded 3-wire power outlet. Report item – opens in a new window or tab.
The manual is divided into eight major sections which provide the following hhp Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a 4935 window or tab.
Manuals, Guides & Specifications for Technicians | Userequip
Output level is Adjusts the transmitter output level continuou;ly displayed on left display. Receiver Board Schematic The A display should be within the range given. When ordered as optionthe A will fit in place of the cover of either the A or A.
Back to home page. Hold Circuit Adjustments Don’t show me this message again. Normal recharge time is about 17 hours. Seller assumes all responsibility for this listing. Keplaceahlc Parts Table Subject to credit approval. Learn more – opens in new window or tab. Sign in to check out Check out as guest.
4935A Transmission Impairment Measurement Set
Page 35 Model Contains line fuse and PC board for selecting line voltage see Section II, paragraph for procedure to select line voltage. The item you’ve selected was not added to your cart.
General Information Model A T a b l e l – l. Tests in this section verify instrument specifications.
HP 4935A Manuals
Page 87 n the Oscilloscope select A vs H operation and connect the probes to the normal input jack. Skip to manuual content. Page 91 Keplaceahlc Parts Table Connect equipment as shown in Figure Intelligent microcomputer control and full alphanumeric display capabilities simplify operation of the A.
GA are described in the following paragraphs. Hewlett-Packard sales and service offices listed at the rear of this manual. Output Level and Transmit F i r s t use a 9OOIl rrsistor, second a !
We have made no changes to this. Cheek the A transmitter at the frcqucncics given in Table Page Title Page Section Section Agilent no longer sells or supports this product.
Delivery times may vary, especially during peak periods. Transmit Monitor Test Output